The Vitalograph copd-6™ BT identifies those at risk of COPD at the pre-symptomatic stage, screening out those whose FEV1 is normal. The screener allows early medical intervention and facilitates better clinical outcomes.
Generates reports via USB connection and Vitalograph Reports software.
One of our 4000 Series of respiratory monitors and screeners, the COPD-6 Monitor comes with:
Get certified device training via our online training resource SpiroTutor™ by Vitalograph® which is available 24/7. We provide one FREE enrollment with the registration of your monitor. A training certificate is issued upon completion.
Product Code: 40451_TH
|Respiratory Monitor COPD-6 USB|
|Dimensions||109mm (length) x 63mm (width) x 42mm (height)|
|Flow Detection Principal||Stator rotor|
|Accuracy||Better than ± 3%|
|Measurement Range||0 – 9.99 L BTPS|
|Power Supply||2 x AAA batteries|
|Battery Life||3 months of use, 3 tests per day|
|Quality Check||Slow start of test (Vext>5%) or a cough detected in the first second|
|FEV substituted for FEV6||When FET < 3s and abrupt end of test|
|Performance Standards||ATS/ERS 2019, ISO 23747:2015, ISO 26782:2009|
|QA/GMP Standards||EN ISO 13485, FDA 21 CFR 820, MDSAP.|
|PC Requirements for Vitalograph Reports||Windows® 7, Windows® 8 or Windows® 10|
|Auto Power Down Time||2 minutes|
|asma-1||asma-1 Child||asma-1 USB/BT||COPD-6||COPD-6 USB/BT||lung monitor||lung monitor USB/BT||lung monitor BT Smart||lung age|
|Import Data to SpV|
A BVF is recommended to use if the copd-6 USB is intended for in clinic multi-subject use. The plastic mouthpiece supplied with the device can be used where the device is intended for single-subject use. For subject groups with excessive sputum production, a BVF can be used to protect the device flowhead from exhaled particulate matter where this may be a concern.
(!) appears after a blow if a cough is detected in the 1st second.
If a new BVF is used for every test and the copd-6 USB is stored in a clean, dust free environment, then the device should require only surface cleaning using a 70% isopropyl alcohol wipe. Full instructions available at Model 4000 – Hygiene & Cleaning Instructions
No, but results can be entered manually.
To connect to a supported mobile App, you would require the lung monitor BT Smart.
For information on any of our products, click the inquiry button, enter your query and we will get back to you.
Would you like to learn more or contact one of our team? If you have a question on any of our products or would like to learn more about our services, please contact us.