The microTM is a handheld spirometer offering full feature functionality. Powered by USB or batteries, the micro is portable and ideal for simple testing and reporting of spirometry.
Ideal for use on subjects within general healthcare and occupational health environments in either clinic based or telemedicine models. Test results and graphs are displayed on the device for instant review. Vitalograph Reports PC reporting software is included as standard.
The flowhead on the micro can be set up to work remotely using the optional Remote Flowhead Kit. Separating the flowhead from the device allows the user to view the device screen during testing.
Keeping the device clean is easy and inexpensive using Bacterial Viral Filters (BVFTM), with validated cross-contamination efficiency >99.999%.
The micro Spirometer comes with:
Product Code: 63310
|Dimensions||83mm x 91mm x 32mm|
|Memory||750 test sessions|
|Flow Detection Principle||Fleisch flow technology|
|Volume Accuracy||Better than ±2.5% or ±0.05L of the reading. (ISO 26782:2009)|
|PEF Accuracy||±10% or ±10L/min of the reading (ISO 23747:2015)|
|Operating Temperature Range||17– 37ºC (ISO26782 limits), Design limits 10 – 40ºC|
|Battery Type||4 x 1.5 AAA alkaline non-rechargeable|
|Power Supply||5V USB power supplied by mini USB cable|
|Safety Standards||EN 60601-1:2006 + A1:2013|
|Performance standards & Guidelines||ATS/ERS 2019, ISO 23747:2015 & ISO 26782:2009|
|Screen||Touchscreen color LCD|
|Printer Options||Use Vitalograph Reports to print to pdf|
|micro||In2itive||In2itive with Spirotrac|
|Total Parameters Available||8 (of 48)||50+||60|
|Quality Measures & Incentives:|
|Data Exchange & Printing:|
|PDF Reports (V=Vitalograph Reports, S=Spirotrac)||V||V||S/V|
|Bi-directional Data Exchange with EMRs|
|Add-On Test Module Options:|
|Resting 12-lead ECG|
The Vitalograph micro uses 4 x 1.5V non-rechargeable IEC60086 certified AAA alkaline batteries. This allows the device to be used without the USB cable connected to the device. Rechargeable batteries may be used but the batteries should be fully charged when commencing testing.
No. Once a test session has ended, a report must be generated to view the results.
For information on any of our products, click the enquiry button, enter your query and we will get back to you.
Would you like to learn more or contact one of our team? If you have a question on any of our products or would like to learn more about our services, please contact us.