Provides accurate and effective monitoring of lung function for respiratory conditions including COPD, cystic fibrosis and post-transplant patients. Ideal for monitoring at home, work, in clinical trials, in primary or secondary care.
The Lung Monitor comes with:
Get certified device training via our online training resource SpiroTutor™ by Vitalograph® which is available 24/7. We provide one FREE enrollment with the registration of your monitor. A training certificate is issued upon completion.
|Respiratory Monitor lung monitor|
|Dimensions||109mm (length) x 63mm (width) x 42mm (height)|
|Flow Detection Principal||Stator rotor|
|Accuracy||Better than ± 3% (FEV1)|
|Back Pressure||Less than 0.15kPa/L/second @ 14L/s|
|Measurement Range||FEV1: 0 – 9.99 L BTPS|
|Maximum Test Duration||6 seconds|
|Quality Check||Slow start of test (Vext>5%) or a cough detected in the first second|
|Power Supply||3V (2 x 1.5V AAA batteries)|
|Expected Battery Life||3 months of use at 3 tests per day|
|Performance Standards & Guidelines||ATS/ERS 2019, ISO 23747:2015, ISO 26782:2009|
|Auto Power Down Time||2 minutes|
|asma-1||asma-1 Child||asma-1 USB/BT||COPD-6||COPD-6 USB/BT||lung monitor||lung monitor USB/BT||lung monitor BT Smart||lung age|
|Import Data to SpV|
A BVF is recommended to use if the lung monitor is intended for in clinic multi-subject use. The plastic mouthpiece supplied with the device can be used where the device is intended for single-subject use. For subject groups with excessive sputum production, a BVF can be used to protect the device flowhead from exhaled particulate matter where this may be a concern.
The (!) appears after a blow if; Vext (extrapolated volume) is > 5% or 100mL of FEV6 or a cough is detected in the 1st second.
If a new BVF is used for every test subject and the lung monitor is stored in a clean, dust free environment, then the device should require only surface cleaning using a 70% isopropyl alcohol wipe. If you suspect that the device flowhead has become contaminated, refer to the 4000 Series Cleaning Instructions.
For information on any of our products, click the inquiry button, enter your query and we will get back to you.
Would you like to learn more or contact one of our team? If you have a question on any of our products or would like to learn more about our services, please contact us.