The Vitalograph asma-1™ is a user friendly electronic respiratory monitor that measures PEF and FEV1 and links easily to other devices by USB for reporting. Ideal for monitoring PEF and FEV1 at home, work, in clinical trials, in primary or secondary care.
It offers greater accuracy than mechanical peak flow meters and eliminates the need for paper record cards.
Vitalograph Reports Software is included and allows you to product PDF reports.
One of our 4000 Series of respiratory monitors and screeners, the asma-1 comes with:
Get certified device training via our online training resource SpiroTutor™ by Vitalograph® which is available 24/7. We provide one FREE enrollment with the registration of your monitor. A training certificate is issued upon completion.
Product Code: 40400_TH
asma-1 USB reports include:
|Respiratory Monitor asma-1 USB|
|Dimensions||109mm (length) x 63mm (width) x 42mm (height)|
|Flow Detection Principal||Stator rotor|
|Back pressure||Less than 0.15kPa/L/second @ 14L/s|
|Measurement Range||PEF: 25 – 840 L/min BTPS FEV1: 0 – 9.99 L BTPS|
|Maximum Test Duration||1 second|
|Quality Check||Time to Peak Flow >120ms or a cough detected in the first second|
|Power Supply||2 x 1.5V AAA batteries|
|Battery Life||3 months of use, 3 tests per day|
|Performance standards:||ATS/ERS 2019, ISO 23747:2015, ISO 26782:2009|
|QA/GMP standards||EN ISO 13485, FDA 21 CFR 820, MDSAP.|
|PC Requirements||Windows® 7, Windows® 8 or Windows® 10|
|Auto Power Down Time||2 minutes|
|asma-1||asma-1 Child||asma-1 USB/BT||COPD-6||COPD-6 USB/BT||lung monitor||lung monitor USB/BT||lung monitor BT Smart||lung age|
|Import Data to SpV|
A BVF is recommended to use if the asma-1 USB is intended for in clinic use. The plastic mouthpiece supplied with the device can be used where the device is intended for single-subject use. For subject groups with excessive sputum production, a BVF can be used to protect the device flowhead from exhaled particulate matter where this may be a concern.
(!) appears after a blow if a cough is detected in the 1st second.
If a new BVF is used for every test and the asma-1 USB is stored in a clean, dust free environment, then the device should require only surface cleaning using a 70% isopropyl alcohol wipe. Full instructions are available at 4000 Series Cleaning Instructions.
For information on any of our products, click the inquiry button, enter your query and we will get back to you.
Would you like to learn more or contact one of our team? If you have a question on any of our products or would like to learn more about our services, please contact us.